New units help to reduce carbon emissions in maternity
31 May 2023
Publish date: 10 October 2022
Air Liquide have informed us that:
1. Specifically for Tandem T:slim insulin pumps with a software version of 6.4.1 or 7.4.3 or lower – the following risks have been identified with these pumps that include a Malfunction 6 alarm (a form of memory corruption); inaccurate/fluctuating battery life display; the touchscreen remaining on indefinitely; and control IQ turning off unexpectedly.
2. The new software (version 7.6) will enable new features including bolus icons displayed on the CGM graph; custom the alarm to resume insulin for instance after stopping for a bath/shower and updates to the options for the cannula fill
3. There is an error in the written Instructions For Use (IFU) leaflet, sent out with Autosoft 90 infusion sets, relating to the cannula fill amount. The correct fill amount for these infusion sets is 0.3 units for the 6mm infusion set and 0.5 units for the 9mm infusion set.
Please check that you are priming your infusion set with the correct amount.
25 May 2023
24 May 2023
12 May 2023
11 May 2023